Real Time Display
Users in VDV
Data Export Task
Setting up a webcam
After install action list
Install VDV on-premise
Creating a new project
Installing scheduled tasks
Restarting the Vista Data Vision System
Exporting site and unit lists from db.robot.c
Exporting the settings from the File Converter
Muting sounds in VDV
Updating VDV with the store in offline mode
Migrate/Move Vista Data Vision to a new server
File Converter gets empty
Update header - new sensors added to the data file
Import modified or historical data
Future data imported into VDV
The VI Not Executable
Email subject contains information from the mail server
Webcam not importing pictures
How to configure Text Message Service
Run tasks minimized
No graphs showing in historical data
Backing up VDV
How to extend VDV Timeout
Updated by Brynhildur
Manual Inclinometer Data Source supports direct input of manually collected readings from inclinometers. Users can upload them through the web interface to add new data to existing sites or create new Manual Inclinometer Sites.
Access the Manual Inclinometer feature under the Data Source menu.
To import new data press the button.
This opens up a place to import data via drag and drop or lets the user select files from the computers directory.
This interface is used to both import new data to existing Manual Inclinometers and also to create new inclinometers.
If the selected data file has not already been imported then a New Manual Inclinometer interface is shown.
Enter a Site Name, select an Owner and make sure to choose the appropriate units and click save. This will create the new Site and import all data from the uploaded file.
When uploading data to a site that has already been created then only new data from the uploaded file is shown.
Under graphs both displacements graphs are now visible (A and B) and a simple click will direct the user to the graph.
For each Manual Inclinometer site the user can edit & delete the site and view the log for all data uploads.
In the Site log a list of all the data measurements is shown. It is possible to delete individual measurements in case a bad reading has been imported.